Potts, P.; Thompson, M. and Wilson, S.
|DOI (Digital Object Identifier) Link:||http://dx.doi.org/10.1111/j.1751-908X.2002.tb00887.x|
|Google Scholar:||Look up in Google Scholar|
Results are presented for round one of a new international proficiency testing programme designed for microprobe laboratories involved in the routine analysis of silicate minerals. The sample used for this round was TB-1, a basaltic glass fused and prepared by the USGS. Thirty nine laboratories contributed data to this round, the majority of major element results being undertaken by EPMA and the majority of trace elements by LA-ICP-MS. Assigned values were derived from the median of results produced by nine selected laboratories that analysed powdered material by conventional ICP-MS, INAA and XRF techniques using bulk powders of the sample. Submitted microprobe results were evaluated using a target precision calculated using the Horwitz function, adopting the same criteria as those used for "applied" geochemistry laboratories in the companion GeoPT proficiency testing programme for laboratories involved in the routine bulk analysis of silicate rocks. An evaluation of results from participating microprobe laboratories indicated that overall, data were compatible with this precision function. A comparison between the performance of bulk and microprobe techniques used in the analysis of the basaltic glass showed remarkably good agreement, with significant bias only observed for the major oxide MgO.
|Item Type:||Journal Article|
|Keywords:||proficiency testing; quality assurance; G-Probe; TB-1; basaltic glass; microprobe laboratories|
|Academic Unit/Department:||Science > Environment, Earth and Ecosystems
|Interdisciplinary Research Centre:||Centre for Earth, Planetary, Space and Astronomical Research (CEPSAR)|
|Depositing User:||Astrid Peterkin|
|Date Deposited:||09 Aug 2007|
|Last Modified:||14 Jan 2016 16:41|
|Share this page:|