Turner, Claire; Knobloch, Henri; Richards, John; Richards, Peter; Mottram, Toby T. F.; Marlin, David and Chambers, Mark A.
PDF (Accepted Manuscript)
- Requires a PDF viewer such as GSview, Xpdf or Adobe Acrobat Reader
|DOI (Digital Object Identifier) Link:||http://doi.org/10.1016/j.biosystemseng.2012.03.001|
|Google Scholar:||Look up in Google Scholar|
Diagnostic tests for some conditions affecting cattle, such as tuberculosis, are often expensive and required over a prolonged period, so that the diagnostic tests involve more than one visit by a qualified vet. An alternative rapid and non-invasive diagnostic test would be desirable. One possibility is the use of breath testing, which has been shown to have diagnostic potential in humans. The development of a device for taking a representative breath sample from a bovine animal is described. Six devices using different configurations were assessed, over three separate testing days, for their ability to take a representative breath sample which does not cause undue stress to the animal and for the ease of operator use. The main factor affecting the sample integrity was dead space, however temperature also played a role. The best samples causing the least stress to animals were taken using a nostril sampler. The nostril samplers were then used to take breath samples from cattle with and without tuberculosis which were then analysed using selected ion flow tube mass spectrometry and gas chromatography-mass spectrometry to demonstrate proof-of-principle.
|Item Type:||Journal Article|
|Copyright Holders:||2012 The Institution of Agricultural Engineers (IAgrE).|
|Project Funding Details:||
|Keywords:||breath analysis; cattle; tuberculosis; SIFT-MS; non-invasive disease diagnosis; VOC monitoring; cattle monitoring|
|Academic Unit/Department:||Science > Life, Health and Chemical Sciences
|Interdisciplinary Research Centre:||Centre for Earth, Planetary, Space and Astronomical Research (CEPSAR)
Biomedical Research Network (BRN)
|Depositing User:||Claire Turner|
|Date Deposited:||03 May 2012 08:38|
|Last Modified:||24 Feb 2016 10:30|
|Share this page:|
Download history for this item
These details should be considered as only a guide to the number of downloads performed manually. Algorithmic methods have been applied in an attempt to remove automated downloads from the displayed statistics but no guarantee can be made as to the accuracy of the figures.