PDF (Version of Record)
- Requires a PDF viewer such as GSview, Xpdf or Adobe Acrobat Reader
Download (8Mb) | Preview
|DOI (Digital Object Identifier) Link:||http://doi.org/10.1088/1748-0221/7/02/C02031|
|Google Scholar:||Look up in Google Scholar|
CCDs have been used in several high resolution soft X-ray spectrometers for both space and terrestrial applications such as the Reflection Grating Spectrometer on XMM-Newton and the Super Advanced X-ray Emission Spectrometer at the Paul Scherrer Institut in Switzerland. However, with their ability to use multiplication gain to amplify signal and suppress readout noise, EM-CCDs are being considered instead of CCDs for future soft X-ray spectrometers. When detecting low energy X-rays, EM-CCDs are able to increase the Signal-to-Noise ratio of the device, making the X-rays much easier to detect. If the signal is also significantly split between neighbouring pixels, the increase in the size of the signal will make complete charge collection and techniques such as centroiding easier to accomplish. However, multiplication gain from an EM-CCD does cause a degradation of the energy resolution of the device and there are questions about how the high field region in an EM-CCD will behave over time in high radiation environments. This paper analyses the possible advantages and disadvantages of using EM-CCDs for high resolution soft X-ray spectroscopy and suggests in which situations using them would not only be possible, but also beneficial to the instrument.
|Item Type:||Journal Article|
|Copyright Holders:||2012 IOP Publishing Ltd and SISSA|
|Extra Information:||Part of 9th International Conference on Position Sensitive Detectors (PSD9), 12–16 September 2011, Aberystwyth, U.K.
Refereed journal article.
|Keywords:||spectrometers; X-ray detectors and telescopes; imaging spectroscopy; avalanche-induced secondary effects|
|Academic Unit/Department:||Science > Physical Sciences
|Interdisciplinary Research Centre:||Centre for Earth, Planetary, Space and Astronomical Research (CEPSAR)|
|Depositing User:||David Hall|
|Date Deposited:||15 Feb 2012 16:33|
|Last Modified:||24 Feb 2016 20:29|
|Share this page:|
Download history for this item
These details should be considered as only a guide to the number of downloads performed manually. Algorithmic methods have been applied in an attempt to remove automated downloads from the displayed statistics but no guarantee can be made as to the accuracy of the figures.