Gow, J.; Murray, N. J.; Holland, A. D.; Burt, D. and Pool, P.
PDF (Accepted Manuscript)
- Requires a PDF viewer such as GSview, Xpdf or Adobe Acrobat Reader
|DOI (Digital Object Identifier) Link:||https://doi.org/10.1117/12.826866|
|Google Scholar:||Look up in Google Scholar|
It has been demonstrated that p-channel charge coupled devices (CCDs) are more radiation hard than conventional n-channel devices as they are not affected by the dominant electron trapping caused by the displacement damage defect the E-centre (phosphorus-vacancy). This paper presents a summary of the results from a comparative study of n-channel and p-channel CCDs each type operated under the same conditions. The CCD tested is the e2v technologies plc CCD47-20, a 1024 ? 1024 frame transfer device with a split output register, fabricated using the same mask to form n-channel and
p-channel devices. The p-channel devices were irradiated to a 10 MeV equivalent proton fluence of 4.07?1010 protons.cm-2 and 1.35?1011 protons.cm-2, an n-channel CCD was irradiated to a 10 MeV equivalent proton fluence of 1.68?109 protons.cm-2, however due to time constraints the n-channel device was not characterised, n-channel comparisons are instead made using a CCD02. As expected the p-channel CCD demonstrated improved radiation tolerance when compared to the n-channel CCD, at -90 ?C there is an approximate ×7 and ×15 improvement in tolerance to radiation induced parallel and serial CTI respectively for equivalent pixel geometries.
|Item Type:||Conference Item|
|Copyright Holders:||SPIE - The International Society for Optical Engineering|
|Keywords:||CCD; proton radiation damage; p-channel; n-channel; charge transfer efficiency; displacement damage hardened;|
|Academic Unit/School:||Faculty of Science, Technology, Engineering and Mathematics (STEM) > Physical Sciences
Faculty of Science, Technology, Engineering and Mathematics (STEM)
|Interdisciplinary Research Centre:||Centre for Earth, Planetary, Space and Astronomical Research (CEPSAR)|
|Depositing User:||Jason Gow|
|Date Deposited:||03 Feb 2010 14:19|
|Last Modified:||29 Nov 2016 17:42|
|Share this page:|
Download history for this item
These details should be considered as only a guide to the number of downloads performed manually. Algorithmic methods have been applied in an attempt to remove automated downloads from the displayed statistics but no guarantee can be made as to the accuracy of the figures.