Murray, N. J.; Holland, A. D.; Smith, D. R.; Gow, J. P.; Pool, P. J. and Burt, D. J.
PDF (Not Set)
- Requires a PDF viewer such as GSview, Xpdf or Adobe Acrobat Reader
|DOI (Digital Object Identifier) Link:||http://doi.org/10.1016/j.nima.2009.01.052|
|Google Scholar:||Look up in Google Scholar|
The CCD247 is the second generation of high-resistivity device to be manufactured in e2v technologies plc development programme. Intended for infrared astronomy, the latest devices are fabricated on high resistivity (~8 kΩ cm) bulk silicon, allowing for a greater device thickness whilst maintaining full depletion when 'thinned' to a thickness of 150 μm. In the case of the front illuminated variant, depletion of up to 300 μm is achievable by applying a gate to substrate potential of up to 120 V, whilst retaining adequate spectral performance. The increased depletion depth of high-resistivity CCDs greatly improves the quantum efficiency (QE) for incident X-ray photons of energies above 5 keV, making such a device beneficial in future X-ray astronomy missions and other applications. Here we describe the experimental setup and present results of X-ray QE measurements taken in the energy range 2-20 keV for a front illuminated CCD247, showing QE in excess of 80% at 10 keV. Results for the first generation CCD217 and swept-charge device (1500 Ω cm epitaxial silicon) are also presented.
|Item Type:||Journal Article|
|Copyright Holders:||2009 ElsevierB.V.|
|Keywords:||X-ray; CCD; SCD; quantum efficiency; QE; full depletion; deep depletion; high resistivity; high-rho; SiLi|
|Academic Unit/Department:||Faculty of Science, Technology, Engineering and Mathematics (STEM) > Physical Sciences
Faculty of Science, Technology, Engineering and Mathematics (STEM)
|Interdisciplinary Research Centre:||Centre for Earth, Planetary, Space and Astronomical Research (CEPSAR)|
|Depositing User:||Neil Murray|
|Date Deposited:||10 Jun 2009 14:50|
|Last Modified:||03 Aug 2016 13:28|
|Share this page:|
Download history for this item
These details should be considered as only a guide to the number of downloads performed manually. Algorithmic methods have been applied in an attempt to remove automated downloads from the displayed statistics but no guarantee can be made as to the accuracy of the figures.